Dam Intake Measurement
Flowing Waters - Power Plants
- Rectangular channel 1.80 x 3.00 m (h x w), partially concrete, partially dry-stone walls
- Part filled
- Rainwater and surface water
- Short calming section
- Manual detection of flow rates (level reading and Q/h-table) destroyed after flood
Definition of tasks
- Continuous, automated flow measurement of turbine intake
- High measurement accuracy despite short and unfavourable calming section for optimum use of the turbine for energy production
- Maintenance-free system required since no personnel available
- Ultrasonic transit time measurement with NivuFlow 650 was chosen due to the relatively clean medium.
- A section of the channel bottom was planed with concrete improving the hydraulic conditions at the measurement point.
- To achieve high accuracy after compensating skew flow behind the bend a crosswise 2-path measurement with transit time sensors mounted in different heights were installed.
Easy installation, easy accessibility, high accuracy, reliability and maintenance-free.