We will keep you informed.
Find out all you need to know about NIVUS - product news, special promotions, practical examples and much more
CS2 flow velocity sensors are suitable for continuous measurement of slight to heavily polluted media.
The pipe-shaped sensors are designed for installation in pipes from the outside using a nozzle. Combined with transmitter types such as NivuFlow 750 and an external level measurement the result is a complete flow measurement system.
|Measurement principle||Ultrasonic Cross Correlation with digital pattern detection (flow velocity)|
Ultrasonic transit time (level measurement)
|Zero point drift||absolutely stable zero point|
|Sensor types||v-sensor with T-compensation|
|Ex-Approval||II 2 G Ex ib IIB T4 Gb (ATEX)|
Ex ib IIB T4 Gb (IECEX)
|Operating temperature||-20 °C to +50 °C|
-20 °C to +40 °C if using the sensors in Ex Zone 1
|Storage temperature||-30 °C to +70 °C|
|Operating pressure||max. 4 bar (for combi sensors with pressure measuring cell max.1 bar)|
|Cable lengths||10/15/20/30/50/100 m pre-configured, can be extended upon request|
|Cable types||LiYC11Y 2 x 1,5 + 1 x 2 x 0,34 PA 1,5/2,5|
|Outside cable diameter||8.4 mm ±0.25 mm|
|Sensor connection||Cable end pre-configured for connection to OCM CF|
|Medium-contacting materials||Polyurethane, stainless steel 1.4571, PPO GF30, PTFE, PA6 GF30|
Optional sensor resistant to chemicals made of PEEK, sleeve made of Hastelloy C-276; cable with FEP coating
|Construction||- Wedge sensor for installation on channel bottom or channel wall|
- Pipe sensor for installation in pipes with sensor screw joint and retaining element
|Measurement uncertainty||for 0.05≤|V|≤0.5 m/s: E_v=0.5/|V| [%]|
for |V|>0,5 m/s: E_v=1 %
Specifications subject to change.
Technical instructions for correlation sensors and external electronic box
Installation instruction for save installation
EC type examination certificate sensor type POA incl. supplement
Ultraschall-Aktivsensoren POA / OCL / CS2
"Ex" - Ultraschall-Aktivsensoren POA / OCL / CS2